Proceedings PaperNonsampling and minimum sampling infrared gas analysis
|Format||Member Price||Non-Member Price|
|GOOD NEWS! Your organization subscribes to the SPIE Digital Library. You may be able to download this paper for free.||Check Access|
Infrared gas analysis has been in use as an industrial process for decades. Normally the gas is sampled from the stack and conveyed to an analyser system. As this requires the gas to be extracted, filtered and cooled to remove condensates, other methods have been developed. These include across-stack gas analysis where an infrared transmitter is placed on one side of the stack and a receiver on the other side. As there is now no need to remove gas from the stack, there are advantages of simplicity and avoidance of loss of soluble gases with the condensate. However, it is not possible to do span and zero tests with the across-stack gas analyser as the measuring path is open. A method which combines the advantages of across-stack and extractive gas analysis is the in-situ gas analyser. In this instrument, a probe is in the stack which fills with stack gas because it is made out of a porous material. This probe has an infrared beam of suitable wavelengths inside. The high quality rigid path compensates for the fact that it is a shorter path than the across-stack gas analyser. The overriding advantage is that zero and span gases can be introduced as they can flush out the porous sampling tube. However, all the advantages of non-sampling are maintained and this method allows the accurate analysis of soluble gases such as NO2, SO2 and NH3.